The Agilent 1200 HiP ALS SL is designed for use with other modules of the Agilent 1200 Series LC system or with other LC systems if adequate remote control inputs and outputs are available.
G1367C HiP-ALS SL
The autosampler is controlled from the Agilent 1200 Series control module or from the Agilent ChemStation for LC systems. The specially-designed thermostattable sample trays holds either 100 × 1.8 ml vials or two wellplates and 10 × 1.8 ml vials. The autosampler thermostat contains Peltier-controlled heat exchangers. A fan draws air from the area above the sample vial tray of the autosampler and is then blown through the fins of the cooling/heating module. There it is cooled or heated according to the temperature setting. The thermostatted air enters the autosampler through a recess underneath the specially-designed sample tray. The air is then distributed evenly through the sample tray ensuring effective temperature control, regardless of how many vials are in the tray. In cooling mode condensation is generated on the cooled side of the Peltier elements. This condensed water is safely guided into the leak system.
This used Agilent G1367C HiP-ALS SL is in great shape. It has been tested on working according factory settings by certified service engineers. The instrument is cleaned outside.
|GLP features||Early maintenance feedback (EMF), electronic records of maintenance and errors|
|Communications||Controller-area network (CAN). RS232C, APG-remote standard, optional four external contact closures and BCD vial number output|
|Safety features||0.1 – 5000 µL in 0.1 µL increments. Recommended injection range 10 – 5000 µL|
|Injection range||Max. loop size 5 ml (selectable range: 5 µL – 50000 µL)|
|Precision||Typically < 0.5% RSD (50 µL loop, overfill factor 3 or greater)|
|Pressure range||Typically < 1.0% RSD (5000 µL loop, 1000 µL injection volume, draw/eject speed 20 mL/min)|
|Sample viscosity range||0.2 – 5 cp|
|Sample capacity||2 well-plates (96 and 384) plus 10 additional 2-mL vials 108 x 2-mL vials in 2 x 54 vial plate plus 10 additional 2-mL vials 30 x 6-mL vials in 2 x 15 vial plate plus 10 additional 2-mL vials 54 Eppendorf tubes (0.5/1.5/2.0 mL) in 2 x 27 Eppendorf tube plate Vial septa and closing mats must be pre-slit.|
|Injection cycle time||< 0.1% with rinse factor > 3 (see test method listing in manual)|
|Carry-over||Typically < 0.01 % using the following conditions: Column: 125 x 4 mm Hypersil ODS, 5 µm Mobile phase: Water/Acetonitrile = 80/20 Flow rate: 1 ml/min Injection volume: 1 µl caffeine (1 mg/ml), 5 µl water to test carryover Outside wash of needle before injection: 20 sec with water using flush port|